Close
Search
Search
Measurement for our planet

- Home
- Research
- Products and services
- Industries
- National challenges
- Our work
- About us
- Strategic partners
- News & Press Office
- Events
- Resources
- Skills and learning
- Careers
- Find us
- Contact us

Close
## M4R has helped more than 400 UK companies accelerate innovation and bring confidence to decision making and investment.

How can we help you?

How can we help you?

We use necessiary cookies to make our site work. We'd also like to set optional analytics cookies to help us improve it. We won't set optional cookies unless you enable them. Using this tool will set a cookie on your device to remember your preferences. We do not use cookies for marketing purposes.

Necessary cookies enable core functionality such as security, network management, and accessibility. You may disable these by changing your browser settings, but this may affect how the website functions.

cookies

We utilise Google Analytics cookies to help us to improve our website by collecting information on how it's used. The cookies collect information in a way that does not directly identify anyone.

- Home
- Resources
- Questions and answers
- How do I convert between dew point and relative humidity?

Questions and answers
# How do I convert between dew point and relative humidity?

**The dew point, **or **dew-point temperature**, is the temperature at which dew, or condensation, forms as you cool a gas. Where the condensate is ice, this is known as the frost point.

**The relative humidity** is the ratio of the amount of water vapour, *e*, in the air to the amount of water vapour, *e _{s}*, that would be in the air if saturated at the same temperature and pressure. It can be expressed

relative humidity (in %) = *e*/*e _{s}* × 100 (Equation 1)

Unfortunately, there is no simple, direct formula for converting between dew point and relative humidity. Conversions between these two parameters must be carried out via the intermediate step of evaluating both the actual vapour pressure of water and the saturation vapour pressure at the prevailing temperature.

- Convert dew-point temperature and ambient temperature into water vapour pressures using Equation 2 or 3 below (or Equation 4 or 5 for greater accuracy)
- Use these values of vapour pressure in Equation 1 to find relative humidity

- Use Equation 2 or 3 below (or Equation 4 or 5 for greater accuracy) to find the saturation vapour pressure from ambient temperature
- Use Equation 1 to calculate water vapour pressure from saturation vapour pressure and known relative humidity
- Use Equation 2 or 3 below (or Equation 4 or 5) to calculate dew or frost point temperature from vapour pressure (requires iteration if using Equations 4 or 5).

Vapour pressure can be calculated using the **Magnus formula.** This states that at a temperature *t* (in °C), the saturation vapour pressure *e _{w}(t)*, in pascals (Pa), over liquid water, is

ln *e _{w}(t)* = ln 611.2 + (17.62

For information, 100 Pa = 1 millibar (mbar)

For the range -45 °C to +60 °C, values given by this equation have an uncertainty of < ±0.6 % of value, at the 95% confidence level.

Over ice,* e _{i}(t)* is

ln *e _{i}(t)* = ln 611.2 + (22.46

For the range -65 °C to +0.01 °C, values given by this equation have an uncertainty of < ±1.0 % of value, at the 95% confidence level.

This is a more accurate but complex alternative formula for vapour pressure (in pascals) from dew point (in kelvin) for water:

ln *e _{w}(T)* = -6096.9385

and for ice:

ln *e _{i}(T)* = -6024.5282

(Formulae due to Sonntag, 1990, updated from formulae given by Wexler, 1976 and 1977.)

The uncertainties associated with these equations are:

- < 0.01 % of value, for water from 0 °C to +100 °C
- < 0.6 %, for supercooled water below 0 °C down to -50 °C
- < 1.0 % for ice down to -100 °C

at the 95% confidence level.

The accuracy of these calculations depends slightly on the pressure and temperature of the gas concerned. For air near room temperature and atmospheric pressure, the water vapour enhancement factor affects the result by approximately 0.5 % of value.

Our research and measurement solutions support innovation and product development. We work with companies to deliver business advantage and commercial success.

Contact our Customer Services team on **+44 20 8943 7070**

© National Physical Laboratory 2021

National Physical Laboratory | Hampton Road, Teddington, Middlesex, TW11 0LW | Tel: 020 8977 3222