Necessary cookies enable core functionality such as security, network management, and accessibility. You may disable these by changing your browser settings, but this may affect how the website functions.
We utilise Google Analytics cookies to help us to improve our website by collecting information on how it's used. The cookies collect information in a way that does not directly identify anyone.
ISO 10360 strongly recommends that a coordinate measuring machine (CMM) be checked regularly during the periods between periodic re-verification. Depending on the measurement tasks for which the CMM is used, the most relevant of the following commonly used artefacts should be chosen. Some examples are given below:
NPL can calibrate a number of these artefacts including purpose-made test pieces, ball plates, hole plates, ball bars, hole bars and ring gauges.
Diameter of CMM reference spheres: NPL also provide a service for measuring the diameter of CMM reference spheres. CMM reference spheres are used for stylus system qualification or as test spheres when carrying out an ISO 10360 probing test. It is therefore important that the diameter and form of these spheres are known with an associated uncertainty. The diameters of reference spheres are measured by comparison with calibrated gauge blocks of similar size. Both metal and ceramic spheres can be calibrated.
Calibration and measurement capability: The calibration and measurement capability for diameter expressed as an expanded uncertainty (k=2) is given by the formula: 0.07 + (0.0011 × diameter in mm) μm. For roundness the uncertainty is typically 0.01 micrometres
Please contact us so that we can discuss your requirements with you.
Our research and measurement solutions support innovation and product development. We work with companies to deliver business advantage and commercial success. Contact our Customer Services team on +44 20 8943 7070