We use necessary cookies to make our site work. We'd also like to set optional analytics cookies to help us improve it. We won't set optional cookies unless you enable them.
We utilise Google Analytics cookies to help us to improve our website by collecting information on how it's used.
Find out more
Electrometers used for radiotherapy applications should meet the requirements of 'IPEM recommendations on dosemeter systems for use as transfer instruments between the UK primary dosimetry standards laboratory (NPL) and radiotherapy centres', Institute of Physics and Engineering in Medicine (IPEM) 2000, Phys. Med. Biol., 45 pp 2445-57
An electrometer can be submitted on its own for calibration at any time or with an associated secondary standard ionisation chamber during annual calibration periods for therapy level, brachytherapy or protection level instruments.
Turnaround time is approximately two weeks if calibrated alone. Uncertainty of calibration is estimated to be ± 0.2 % (k=2).
Don’t see what you are looking for? Our diverse skill set enables us to provide bespoke solutions. Please contact us to discuss your requirements.
Our research and measurement solutions support innovation and product development. We work with companies to deliver business advantage and commercial success. Contact our Customer Services team on +44 20 8943 7070