Necessary cookies enable core functionality such as security, network management, and accessibility. You may disable these by changing your browser settings, but this may affect how the website functions.
We utilise Google Analytics cookies to help us to improve our website by collecting information on how it's used. The cookies collect information in a way that does not directly identify anyone.
Calibration of an ionisation chamber in terms of absorbed dose to water, based on primary standard electron-beam graphite calorimeter is available in Spring and Autumn each year.
This service is designed to be used with the current IPEM Code of Practice (2003), which is based on this service. The IAEA Technical Report No.398 can also be used, but this report is not recommended by the IPEM.
Calibration of NACP-02, Roos, Roos type chambers (PTW, IBA and Sun Nuclear) and graphite-walled Farmer type chambers (NE2571, PTW30004). This uses R50 alone as beam quality specifier to define reference depth. Beam qualities presently covered in the range R50=1.6 cm to 8 cm (nominal incident electron energies 4 MeV to 20 MeV). Uncertainties for direct calibration ±2 % at the 95 % confidence level. Characterisation of recombination behaviour of each chamber in terms of dose per pulse.
Don’t see what you are looking for? Our diverse skill set enables us to provide bespoke solutions. Please contact us to discuss your requirements.
Our research and measurement solutions support innovation and product development. We work with companies to deliver business advantage and commercial success. Contact our Customer Services team on +44 20 8943 7070