Necessary cookies enable core functionality such as security, network management, and accessibility. You may disable these by changing your browser settings, but this may affect how the website functions.
We utilise Google Analytics cookies to help us to improve our website by collecting information on how it's used. The cookies collect information in a way that does not directly identify anyone.
NPL calibrates the cladding diameter of optical fibres in the range 123 µm to 127 µm. We use a non-contact measurement method and can measure customer supplied fibres or supply a calibrated reference fibre mounted in a retractable fibre holder which is used to protect the calibrated cleaved fibre end.
We follow the IEC reference test method based on: Fibre geometry IEC 60793-1-20 Cladding Mean Diameter Reference Fibre – Video Grey Scale Technique – Annex B.
NPL calibrates the outer coating diameter of optical fibres in the range 230 µm to 260 µm. We use a non-contact measurement method and can measure customer supplied fibres or can supply a calibrated reference fibre, which is usually about 150 mm long. Measurements are typically made 10 mm from one end of the fibre.
We follow the IEC reference test method based: Fibre geometry IEC 60793-1-20 – Coating Diameter – Outer Coating Diameter Reference Fibre – Side View Light Distribution – Method A.
NPL can supply a length of specially designed optical fibre with multi-step refractive index profile. This fibre can be used to calibrate the refractive index difference scale of a customer's profiler. In addition, the intermediate steps can be used to assess non-linearity.
Don’t see what you are looking for? Our diverse skill set enables us to provide bespoke solutions. Please contact us to discuss your requirements.
Our research and measurement solutions support innovation and product development. We work with companies to deliver business advantage and commercial success. Contact our Customer Services team on +44 20 8943 7070